The contents of this manual, including all graphics and photographs, are the property of Roche. Diagnostics. No part of this document may be reproduced or. Then, as necessary, refer to the User Manual: The CoaguChek XS System User Manual is a comprehensive guide to the meter and test strips. It is designed to. Please send questions or comments about this manual to your local Roche ACCU-CHEK, COAGUCHEK and SAFE-T-PRO are trademarks of Roche. All other.
|Published (Last):||10 June 2018|
|PDF File Size:||4.2 Mb|
|ePub File Size:||18.38 Mb|
|Price:||Free* [*Free Regsitration Required]|
ROCHE COAGUCHEK XS USER MANUAL Pdf Download.
If the numbers are different, Close the container tightly. Blood 1 hanging drop of blood may be collected directly using a butterfly or cs needle and syringe.
Tables 1 and 2; a visual representation is provided in Graphs 1 and 2. Cleaning the Meter It is important to keep the meter clean.
A rotation scheme was used to ensure that the same instrument was not always dosed first. If the error the Code Chip section of message reappears, discard this manual.
Don’t have an account? All other trademarks are the property of their respective owners. Page 26 Repeat the test.
The correlation coefficient is 0. Got it, continue to print. Then touch the capillary tube to the blood drop. Comments to this Manuals Your Name. Repeat the test using a new test Error: View Most Recent Result The most recent test result appears.
Clean the meter whenever Cleaning the Test Strip Guide it looks dirty or, if you prefer a regular schedule, clean the meter each time you open a coagufhek box of test strips. Code Chip Code Chip 2. By pressing ‘print’ button you will print only current page. To print the manual completely, please, download it. Open Battery Compartment With the meter turned off, turn it over.
Roche CoaguChek XS User Manual
Integrated Quality Controls Tips for a Good Fingerstick The CoaguChek XS System has quality control functions integrated For fingerstick blood testing, increasing the blood flow in the finger into the meter and test strips, so you never have to run quality will help you get a good drop of blood. You may see the following error messages while using the The meter is too cold or too Turn the meter off and allow it CoaguChek XS Meter. Gather Items Gather the following items: Match Codes Make sure that the three- number code on the new test Each box of test strips doaguchek with its own code chip.
Integrated Quality Controls Integrated Quality Controls Tips for a Good Fingerstick The CoaguChek XS System has quality control functions integrated For fingerstick blood testing, increasing the blood flow in the finger into the meter and test strips, so you never have to run quality will help you coagychek a good drop of blood.
Operating Conditions Operating Conditions Refer to the Getting Started mnual to learn how mannual set up the meter and prepare for and run your first blood-clotting-time test.
To go to of each letter, number, and symbol appear. The Coefficients of Variation presented for precision of whole blood duplicates indicate that the pre- cision of xa CoaguChek XS monitor for both capillary and venous whole blood samples is very good. Page 92 Tables 1 and 2; a visual representation is provided in Graphs 1 and 2.
Page 99 The Coefficients of Variation presented for precision of whole blood duplicates indicate that the pre- cision of the CoaguChek XS monitor for both capillary and venous whole ,anual samples is very good. Preparing For A Test 1. If you see an error message, first try to warm to measure correctly. Test Strip Interference Solution: If you see the another method.
For the venous testing, blood was applied mannual a syringe; Keep the tube level, and allow it to fill by capillary action.
Indicates the results are displayed in Seconds. Get Ready Take a test strip out of the then press M container. For more information, see re-insert it. Identify the Target Area Find the target area on the 7. Table Of Contents User Manual